Plug in your numbers to exchange them for numbers from the most current edition. Elemental cost analysis pdf numbering changes to the new edition. Canadian customers may purchase from the Construction Specifications Canada. Construction Specifications Institute and Construction Specifications Canada.
Type or paste a DOI name into the text box. Some of the features on CT. We hope you find this website both helpful and informative. We sell two devices for injecting gas or liquid into the EA. One would mount as a replacement for the autosampler, and the other is an in-line injector which would usually be installed before the water trap. The in-line injector would be used by someone who only wanted to use the GC part of the EA to get gases into the MS. Do you carry a product called Chromosorb W?
I can’t seem to find it on your website? I have a question about running filter samples on a CN analyser. There is no real problem with GFFs, but you have to pay close attention to the buildup of glass in the reactor. We recommend using the quartz inserts to catch the glass and make it easy to remove from the reactor. If you just drop it into the main combustion reactor it is almost impossible to remove.
The H2O peak will always tail, there is no way to get it to look like either nitrogen or carbon. Dear David,For NCS on the NA 2500, I would recommend using our pre-packed reactors for sulfur, which are packed with Tungstic oxide and long copper wires. These will give you the best peak shape for sulfur. As far as a GC column goes, our two meter GC column for sulfur will separate the N,C, and S peaks. Eurovector for HD and 18O analysis in water. The valve you are describing is called the bi-matic valve, and it was used in older CE analyzers to inject the oxygen for sample combustion. It was a six port switching valve which was air actuated.
There was a viton membrane in the valve which controlled the flow of gases in and out of the valve and it needed to be replaced every few years. The NC 2500 does not have the bi-matic valve. In the NC 2500 electrovalves control the flow of gases and are used to inject the oxygen for combustion. So, you don’t have to worry about replacing the membrane.